Not getting movement from the actuator(s) from my handset but am able to in manual command when using some station configurations.
This can be caused by too small of a Remote Setpoint Null Pace Band. Go to the Station Manager Display Menu > Station Setup > Tuning > Advanced Tab > Remote Setpoint Null Pace Band (Figure 1). The default value is 10% of the full scale. You can start with that and increase it as necessary. After making a change go to the File Menu > Save Parameters.
Figure 1 – Remote Setpoint Null Pace Band Setting